THE CLIENT’S NEEDS
A large pharmaceutical client engaged NOVO to develop automated test equipment (ATE) for the functional testing of a modified drug delivery platform for a wearable device. The equipment was to accommodate eight devices and independently measure injection time, dose accuracy, and efficiency. Another requirement was to verify the operation of the dose delivery indicator LED. In addition to the wearable device testing hardware, the system required a custom software application to log test results and setup information, and to provide a user interface (UI) to guide the operator through the test procedure.
THE ENGINEERING BEHIND GREAT PRODUCTS
The wearable device testing equipment developed for the system included:
- A custom, three-color detector with calibrated wavelength ranges used to establish LED on-off and color states.
- A custom audio output test circuit with a notch filter for the frequency of interest (and rejection of ambient noise).
- An off-the-shelf (OTS), high-accuracy load cell with custom signal processing algorithms applied to the output to enable the differentiation of individual droplet delivery from stream delivery.
- A vision system to image the device cannula and a custom image-processing algorithm to measure cannula extension.
This equipment was tested and verified on site at the client’s facility, where it is used as a design development tool. NOVO’s ability to combine sophisticated software-implemented algorithms to process images or electrical signals with elegant electromechanical hardware solutions makes us well suited to ATE projects for medical device products.